Octoroll® Motor Unit
Simply place a UNIVERSAL OCTOPAQUE® unit on the four wheels of the OCTOROLL®, and the examination is ready to be conducted from the control bay.
Immobilization is safe, efficient and fast. The next baby may even be pre-immobilized outside the room on another OCTOPAQUE®, and brought inside for immediate exchange on the OCTOROLL®.
Upper gastrointestinal series, colon, voiding cystograms, etc. can be done without tedious loss of time.
The radiologist remains protected from radiation, yet just fingertip away from the patient, while assistants supervise from a safe distance.
The unit is totally wireless since it is battery operated and controlled by radio signals. The battery charge is sufficient for an average day’s work, which may even continue while recharging. Stops are instantaneous and a brake keeps the baby perfectly still during exposures.
Light and compact, the unit is put away on a shelf under continuous charge for immediate use at any time.
Adapts instantly to any OCTOPAQUE® unit and to any x-ray apparatus
Specifications OCTOROLL® UM-02 (Approved by the FDA)
Motor module :
Motors: two 13-commutator dc of the highest quality. Instant starts and stops.
Gears: two worm type. Provide strong internal brake when not rotating..
Frame: strong aluminum made with computer precision.
Battery: sealed lead acid, 12 volts (supplied), with a life expectancy of over three years..
Battery controller: 13.8 volts dc, with blinking red led at 50% discharge, indicating need for recharge, to prolong battery life.
Radio receiver: rf frequencies with 256 safety codes. Resist local incoming interference, which must nevertheless be accepted.
Switches: a) main with led.. B) local rotation switch, dead man type, clockwise and counterclockwise, overrides remote control module.
Maximum weight: 16 kg. Or up to 25 kg with optional wider and stronger board.
Dimensions: 37 l x 18 h x 16.5 w cm; 4 kg; feet component: 28 l x 10.5 h. X 10 w cm; 800 gm.
Remote control module :
Battery: 9 v. Alkaline (supplied), good for over one year. A buzzer in the motor module warns when battery is low (7 volts).
Switches: two dead man switches, for clockwise and counterclockwise rotation. Buttons are recessed to prevent accidental operation. Simultaneous activation of both switches causes a buzz, but no action or damages.
Compliance: with fcc rules part 15. Possible outgoing local interference must be accepted in a 235 m. Radius; verification must be made before operation and modification of security codes done if necessary.
Type: wall plug, csa and ul approved. Rated 800 ma.
Input: 110 v. Ac. Output: 15 to 17 volts ac. When connected, allows operation of the unit.
220 volts adapter is available as an option.
Output: 15 to 17 volts ac. When connected, allows operation of the unit.
*subject to change without notice.
** examinations can always be immediately conducted manually by installing the immobilized baby directly on the x-ray table
***this device is not to be used in a magnetic resonance environment.
OCTOSTOP® Inc. and its personnel do not assume any liability regarding the use, indications, consequences, or any situation directly or indirectly related to its products.